Heated Probe Tip
Contains a heating element which is heated to 100°C (212°F) on the default setting for the destruction of human tissue. The probe tip should only be applied to tissue which is intended to be ablated. The TC probe shaft is not intended to deliver heat. The probe shaft and sidewall tip has a maximum temperature of 43°C (109°F). Contact with the probe shaft should be avoided during the procedure.
Automated procedure timing with visual and audible indicators – no watch or clock required. Can be set for 20, 30, 40, or 60 seconds and 80, 90, 100, or 120°C.
Heat quickly (5-8 seconds) that can be autoclaved for sterilization.
Lightweight with no tether wires or cables required to connect probe to device and no external power supply required for treatment.
- Power Supply: 11.1 VDC
- Battery Pack: Rechargeable Lithium-Ion, 3-cell 2AH Battery Pack, BMS overcharge protection
- Battery Charger: 100 – 240 VAC, 1.0A, 50-60Hz Input; 12.6VDC, 1.8 amp Output
- Charge Time: Two(2) hours
- Full-Charge Activation: Until low-battery indicator illuminates
- Power Output: 30 Watts
- Treatment (Duty) Cycle: ~8 seconds of heat up, 20 seconds of therapy at 100°C, and ~10 seconds of cool down
- Power Output: 30 Watts
Dimensions and Weight
- Width: 1.5 inches (4 cm)
- Height: 8 inches (20 cm)
- Depth: 2 inches (5 cm)
- Weight: 11 oz (240 g)
- Ambient Temperature: 10 to 40 degrees Celsius
- Relative Humidity: 0% to 80% non-condensing
Transport and Storage
- Ambient Temperature: -40 to 70 degrees Celsius
- Relative Humidity: 0% to 80% non-condensing
- Type B Applied Parts
- IP21 Rating: Solid particle protection: Level 2 (>12.5mm), Liquid ingress protection: Level 1 (dripping water)
- 19mm Flat probe (2x)
- 19mm Nipple probe (1x)
- 16mm Flat probe (1x)
- Batteries (2x)
- Battery charger (1x)
- Carrying case (1x)
Product Documents, Manuals & Brochures
Battery Data Sheet
ISO 13485 Certificate
How do I know that thermocoagulation™ is as effective as cryogenic therapy?
Literature studies have validated that the outcome clinical efficacy is as good or better with thermocoagulation™ as compared to freezing. Refer to Duncan’s studies as shown in Dolman’s meta-analysis in the website educational section. The Liger thermocoagulator™ uses the latest state-of-the-art technology to reach and maintain the treatment tip accurately at temperature within 0.2°C.
What is cold coagulation as compared to thermocoagulation™?
How long and at what temperature should pre-cancerous cervical lesions be treated?
How many treatments can I expect with a full battery charge?
How do I clean the Probe?
The full high-level sterilization is given in the IFU that was sent with each unit on page 5 column 2. Many users follow the method below for cleaning and Hi-level disinfection of the probes:
Place the silicon cap firmly over the connector end. Clean any debris off the probe with mild detergent and water using a soft cloth or soft brush, then rinse in clean water.
Place the probes upright in a cup (approximately 5- 6” deep) of ~2.5% gluteraldeyde (Cydex) for 20-60 minutes (some sites use a 0.5% Chlorine solution-Chlorox).
This allows the connector end not to be immersed in solution which is not necessary as the connector end is not inserted into the patient nor contacts the patient and need not be sterilized.
Some users apply a heat cycle of the device to assure that the heated end has reached a sterilization temperature but this is unnecessary and regulatory agencies have not approved this procedure for sterilization. It certainly does sterilize the tip as all viral proteins are denatured above 60’C.
Rinse the probe end (not the connector end with the silicone cap) with clean, non-contaminated water, thoroughly dry the probe in air.
Wrap in clean material to keep it from being contaminated before next use.
The handle can, if needed, be wiped clean with a cloth dampen with a small amount of alcohol.
Is there any difference between the two batteries in the pack?
What is the depth of tissue destruction for a 20 second/100°C cycle?
How does the thermocoagulator™ compare to cryotherapy?
The thermocoagulator™ is quicker, easier and simpler to use. The thermocoagulator™ procedure is approximately 20 seconds per treatment as compared to cryotherapy which requires 11 minutes per treatment. The thermocoagulator™ has easy to use rechargeable batteries that use minimal energy, whereas cryogenic requires costly replacement gas cylinders.
- 1x Thermocoagulator Main Unit including 1x HTU-110 handle
- 2x 19mm Flat probe
- 1x 19mm Nipple probe
- 1x 16mm Flat probe
- 2x Rechargeable lithium ion batteries
- 1x Charger
- 1x Instruction manual
- 1x Carrying case
Liger warrants each product manufactured by it to be free from defects in material and workmanship under normal use and service for the period(s) set forth below.
Liger’s obligation under this warranty is limited to the repair or replacement, as its sole option, of any product, or part thereof, which has been returned to it or its Distributor within the applicable time period shown below, after delivery of the product to the original purchaser, and which examination discloses, to Liger’s satisfaction, that the product is indeed, defective.
This warranty does not apply to any product, or part thereof, which has been repaired or altered outside Liger’s factory in a way so as, in Liger’s judgment, to affect its stability or reliability, or which has been subjected to misuse, neglect, or accident.
This warranty is in lieu of all other warranties, express or implied, including without limitation, the warranties of merchantability and fitness for a particular purpose, and of all other obligations or liabilities on the part of Liger.
Liger neither assumes nor authorizes any other person to assume for it any other liability in connection with the sale or use of any of Liger’s products.
Notwithstanding any other provision herein or in any other document or communication, Liger’s liability with respect to this agreement and products sold hereunder shall be limited to the aggregate purchase price for the goods sold by Liger to the customer.
Liger disclaims any liability hereunder or elsewhere in connection with the sale of this product, for indirect or consequential damages.
This warranty and the rights and obligations hereunder shall be construed under and governed by the laws of the State of Utah, United States of America (USA). The sole forum for resolving disputes arising under or relating in any way to this warranty is the 3rd District Court of Utah, USA. Liger, its dealers, and representatives reserve the right to make changes in equipment built and/or sold by them at any time without incurring any obligation to make the same or similar changes on equipment previously built and/or sold by them.
Smyta Joshi, MD Hirabai Hospital Pune, India
Dr. Kofi Effah , Cervical Cancer Prevention and Training Centre, Catholic Hospital, Battor, Ghana
Beatrice Kabota, Nkhoma CCAP Hospital, Lilongwe Malawi